James Wren (Prosig UK) explains how to store FRFs, coherence and/or time series data in modal hammer testing using Prosig’s P8000 & DATS software.
3263class="post-3263 post type-post status-publish format-video hentry category-dats category-tutorial category-video tag-components tag-dats tag-hammer tag-hammer-impact tag-hammer-testing tag-james-wren tag-modal tag-modal-analysis-software tag-modal-testing tag-nodes tag-p8000 tag-prosig tag-structures post_format-post-format-video"
James Wren (Prosig UK) explains how nodes, components & structures are used in modal hammer testing using Prosig’s P8000 & DATS software.
3250class="post-3250 post type-post status-publish format-standard hentry category-p5000 category-tutorial tag-accelerometer tag-accelerometers tag-calibration tag-dc-level tag-electrical-conductance tag-prosig tag-sensitivity-change tag-shaker tag-signal-quality tag-sine-wave tag-transducer tag-vibration"
Here we look at how to calibrate an accelerometer using a Prosig P5000 system.
Put the wax on the shaker top. Place the accelerometer in the axis you wish to calibrate with positive up and cable connected to P5000 with the relevant transducer class chosen.
Go to Single Channel Calibration screen.
Click on the Tone tab.
With P5000 armed turn on the shaker and monitor the sine wave on the real-time monitor and check Signal Quality as being GOOD.
With a GOOD sine wave click the Calculate button. It is recommended to click Calculate three times.
Check to see the Sensitivity change from what was originally entered when setting up the channel transducer information to the new calculated value. The calculated value should be close to the original.
DC Cal Offset is for DC level accelerometers.
Requested Excitation is for non IEPE (ICP) accelerometers such as capacitive.
After calculating the new sensitivity click on the Use button to make the change in the Transducer Sensitivity Acquisition Setup file.
NOTE: To achieve a GOOD signal reading place the shaker on a flat surface and avoid touching during operation. Make sure the surface the shaker is sitting on does not come in contact with other sources of vibrations or electrical conductance.3257class="post-3257 post type-post status-publish format-video hentry category-dats category-p8000 category-tutorial category-video tag-data-acquisition-hardware tag-hammer tag-hammer-impact-test tag-hammer-test tag-impact-test tag-modal-analysis tag-modal-analysis-software tag-p8000 tag-prosig tag-structural-testing tag-vibration post_format-post-format-video"
James Wren (Prosig UK) provides a step-by-step guide to performing a Hammer Impact Test on a structure using Prosig’s DATS software and P8000 data acquisition hardware.
3242class="post-3242 post type-post status-publish format-video hentry category-p8000 category-tutorial category-video tag-calibrate tag-calibration tag-measurement tag-prosig tag-strain tag-strain-gauge tag-test tag-video post_format-post-format-video"
James Wren (Prosig UK) steps through how to calibrate a strain gauge for use with a Prosig P8000 system.
3011class="post-3011 post type-post status-publish format-standard hentry category-p8000 tag-dc-component tag-dc-components tag-dc-source tag-high-pass-filter tag-iepe tag-prosig tag-sensors tag-sine-wave tag-volts tag-zero-function"
The Prosig P8000 Auto-Zero function is used to remove the DC content of a signal.
For example, if you have a sine wave that is ±5 volts about 0 volts it has no DC offset. But if you have a sine wave with ±5 volt amplitude about 3 volts, then it has a 3 volt DC offset. Below is an example of these two sinewaves.
Continue reading Why does Auto-Zero have no effect for IEPE sensors?2990class="post-2990 post type-post status-publish format-status hentry category-news tag-ford-motor tag-ford-motor-company tag-kindle tag-kindle-touch tag-prize tag-prosig tag-user-survey post_format-post-format-status"
Thank you to everyone who completed our 2012 User Survey. The results were, as always, fascinating and will be used to improve our future hardware & software products.
We are please to announce that James Hansen of Ford Motor Company was the winner of the prize draw for a Kindle Touch. James is a veteran of Prosig hardware & software and has seen several generations of both. Here is James receiving his Kindle from Ken Goward of Prosig USA…
Congratulations to James. We hope you enjoy your Kindle.
Bad luck to all our other respondents. There’s always next year’s survey…2974class="post-2974 post type-post status-publish format-standard hentry category-p8000 tag-basic-circuit-theory tag-cable-length tag-cable-type tag-electrical-conductor tag-frequency-bandwidth tag-high-frequencies tag-p8000 tag-phenomena tag-prosig tag-rf tag-wavelength"
Using 50Ω or 75Ω cable will not have any negative effects on the P8000 at all. (Generally speaking, only large voltages will damage the P8000). The impedance of the cable can only ever effect the signal the cable is carrying. Continue reading Should I use 50 Ohm or 75 Ohm cables with a Prosig P8000 system?2909class="post-2909 post type-post status-publish format-standard hentry category-dats category-signal-processing category-tutorial tag-acceleration tag-animation-software tag-frf tag-modal-analysis-software tag-natural-frequencies tag-operating-deflection-shapes tag-prosig tag-spectra tag-vibration tag-wire-frame-model"
ODS = Motion of structure vibrating naturally (for example bridge vibration) or when excited by an unknown force (for example generator vibration). Prosig’s Structural Animation software performs ODS analysis.
OMA = Modal Analysis
Prosig’s Modal Analysis software performs OMA. Continue reading What is the difference between Operating Deflection Shapes (ODS) and Operational Modal Analysis (OMA)?
One of Prosig’s engineers was recently creating a complex worksheet application that required some complicated arithmetic to be performed on values calculated in the worksheet. Worksheet (or global) variables are ideal for this. The Variable Mods DPU also has an expression evaluator built in. Consider the following worksheet that takes a Named Element from two signals and sums them. Continue reading Using global variables, variable arithmetic & script fragmentsProudly powered byWordPressDesign byBlogoLife