lang="en-US"UTF-8 accelerometer Archives - Prosig Support Blog class="archive tag tag-accelerometer tag-51 custom-background custom-background-image group-blog blogolife-3_0_7 unknown"

# Prosig Support Blog

## The place to come for support for Prosig's DATS, P8000 & PROTOR products

3462class="post-3462 post type-post status-publish format-standard hentry category-signal-processing category-tutorial tag-accelerometer tag-conversion tag-convert tag-db tag-decibels tag-linear tag-log tag-microphone tag-power-spectrum"

## How Do I Convert To Decibels?

fa fa-file-text-o

Much confusion revolves around linear and non-linear numbers. The following outlines the mathematical process to convert from a number expressed in dB to a linear quantity. How do we convert to decibels and back again? Continue reading How Do I Convert To Decibels?

Add a Comment3385class="post-3385 post type-post status-publish format-standard hentry category-tutorial tag-accelerometer tag-accelerometers tag-bridge tag-charge tag-dc tag-high-pass-filter tag-iepe tag-wheatstone-bridge"

## What Type of Accelerometer Should I Use? IEPE, Charge or Bridge-based?

fa fa-file-text-o

#### What type of accelerometer should I use? What are the advantages/disadvantages of a charge mode accelerometer, an IEPE accelerometer and a bridge based accelerometer?

There are so many types of accelerometer that is often difficult to know what type of accelerometer to use. An IEPE accelerometer will have a high pass filter at about 5Hz. The charge type will, by it’s nature have what is effectively a high pass filter at about 0.1Hz. Therefore neither type will show DC levels. The charge type will usually have a lower frequency bandwidth than the IEPE type. Charge accelerometers can be used at higher temperatures however. Continue reading What Type of Accelerometer Should I Use? IEPE, Charge or Bridge-based?

Add a Comment3294class="post-3294 post type-post status-publish format-standard hentry category-signal-processing category-tutorial tag-acceleration tag-accelerometer tag-accelerometers tag-acquisition tag-hammer-impact tag-hammer-impact-test tag-hammer-test tag-inlet-manifold tag-manifold tag-resonance tag-resonant tag-rotating-machinery tag-rotation-speed"

## How do I find the natural frequency of an inlet manifold?

fa fa-file-text-o

There are a number of ways to find the natural frequency (resonance) of a part like an automotive inlet manifold. Here are three different types of popular test technique. But which one should you use and why? Continue reading How do I find the natural frequency of an inlet manifold?

Add a Comment3285class="post-3285 post type-post status-publish format-standard hentry category-p8000 category-signal-processing category-tutorial tag-accelerometer tag-cylinder-head tag-engine tag-inlet-manifold tag-manifold tag-noise tag-tachometer tag-transducer tag-vibration"

## How do I perform vibration analysis on a cylinder head and inlet manifold?

fa fa-file-text-o

We were asked the following question…

I want to perform some cylinder head and inlet manifold vibration analysis, what should I do?

First we need to consider sensor selection Continue reading How do I perform vibration analysis on a cylinder head and inlet manifold?

Add a Comment3250class="post-3250 post type-post status-publish format-standard hentry category-p5000 category-tutorial tag-accelerometer tag-accelerometers tag-calibration tag-dc-level tag-electrical-conductance tag-prosig tag-sensitivity-change tag-shaker tag-signal-quality tag-sine-wave tag-transducer tag-vibration"

## Calibrating an Accelerometer with a Prosig P5000 system

fa fa-file-text-o

Here we look at how to calibrate an accelerometer using a Prosig P5000 system.

Put the wax on the shaker top. Place the accelerometer in the axis you wish to calibrate with positive up and cable connected to P5000 with the relevant transducer class chosen.

Go to Single Channel Calibration screen.

Click on the Tone tab.

With P5000 armed turn on the shaker and monitor the sine wave on the real-time monitor and check Signal Quality as being GOOD.

With a GOOD sine wave click the Calculate button. It is recommended to click Calculate three times.

Check to see the Sensitivity change from what was originally entered when setting up the channel transducer information to the new calculated value. The calculated value should be close to the original.

DC Cal Offset is for DC level accelerometers.

Requested Excitation is for non IEPE (ICP) accelerometers such as capacitive.

After calculating the new sensitivity click on the Use button to make the change in the Transducer Sensitivity Acquisition Setup file.

NOTE: To achieve a GOOD signal reading place the shaker on a flat surface and avoid touching during operation. Make sure the surface the shaker is sitting on does not come in contact with other sources of vibrations or electrical conductance.

Add a Comment2968class="post-2968 post type-post status-publish format-standard hentry category-dats category-tutorial tag-accelerometer tag-accelerometers tag-angles tag-balance tag-balancing tag-centre-of-gravity tag-measurement tag-multi-plane-balancing tag-sensors tag-shaft-balancing tag-shaft-bearing tag-speed-range tag-tachometer tag-vertical-planes tag-vibration tag-vibrations"

## How do I balance a shaft?

fa fa-file-text-o

“How do I balance a shaft?” seems like a fairly straightforward question, but there are a number of things that we need to understand first. Here we look at a number of key concepts that need to be understood in order perform balancing.

### What does balance mean?

Well broadly speaking to balance a shaft, mass must be added or removed at certain angles. The concept being that the centre of gravity and rotational centre of the shaft will be equal when the shaft is balanced. Continue reading How do I balance a shaft?

1 Comment3058class="post-3058 post type-post status-publish format-status hentry category-dats category-news category-p8000 tag-accelerometer tag-accelerometers tag-animation-packages tag-hammer tag-hammer-impact tag-impact-software tag-modal-analysis tag-new-features tag-new-software tag-preview-version tag-report-generation tag-software-preview tag-vibration post_format-post-format-status"

## New Hammer Impact Software – Preview coming soon

fa fa-file-text-o

After listening to customer feedback, we have completely re-engineered the DATS Hammer Impact software. The new version will ship with the next DATS update (not the soon-to-be-released V7.0.23). In the meantime you will be able to download and preview the new package. You will of course need a P8000 system and a DATS licence with the Hammer Impact option enabled. Continue reading New Hammer Impact Software – Preview coming soon

Add a Comment2763class="post-2763 post type-post status-publish format-standard hentry category-dats category-tutorial tag-accelerometer tag-accelerometers tag-dats tag-example-worksheet tag-resultant tag-time-histories tag-time-history tag-triaxial tag-waterfall tag-waterfalls"

## How Do I Get An XYZ Resultant Waterfall

fa fa-file-text-o

A user has three signals captured using a triaxial accelerometer and asked “What is the simplest way to get the XYZ resultant from run-up file?” He had tried forming a resultant of the raw time histories, but didn’t fully understand the resultant time history.

Of course, the correct way of processing the data is to calculate the individual waterfalls from the x, y & z data and then calculate a resultant waterfall. Continue reading How Do I Get An XYZ Resultant Waterfall

Add a Comment2898class="post-2898 post type-post status-publish format-standard hentry category-dats category-signal-processing category-tutorial tag-accelerometer tag-accelerometers tag-engine-noise tag-microphone tag-vibration tag-wheel-hub"

## What is Source Contribution Analysis (or SCA)?

fa fa-file-text-o

How do you measure the causes of a noise or vibration with respect to several sources? Which source is causing what part of the response?

For example, how does the noise inside the cabin of a vehicle relate to the engine noise or wheel hub noise and vibration? Continue reading What is Source Contribution Analysis (or SCA)?

Add a Comment2815class="post-2815 post type-post status-publish format-standard hentry category-dats category-p8000 category-signal-processing category-tutorial tag-acceleration tag-accelerometer tag-bearing-analysis tag-bearings tag-dats tag-frequency-range tag-hall-effect-sensors tag-infra-red-sensors tag-lasers tag-maximum-frequency tag-rollers tag-rotary-position tag-rotation-speed tag-shaft-speed tag-tachometer tag-test-specifications tag-vibration-level"

## A Simple Step-by-step Guide To Bearing Vibration Analysis

fa fa-file-text-o

Generally. when developing and testing bearings a simple step by step procedure should be followed.