The place to come for support for Prosig's DATS, P8000 & PROTOR
James Wren is an Application Engineer and the Sales Manager for Prosig Limited. James graduated from Portsmouth University in 2001, with a Masters degree in Electronic Engineering. He is a member of the Institution of Engineering and Technology. He has been involved with motorsport from a very early age with special interest in data acquisition. James is a founder member of the Dalmeny Racing team.
When using an analogue channel input channel of a P8000, with a tachometer sensor, is it possible to utilise the tachometer processing functions as you would if you were using a tachometer input of the P8000?
For example could you display an RPM on an analogue channel?
When using an analogue channel it is possible to select, from the signal setup matrix, ‘Trigger Type’, with this option set to ‘Tacho’ and the ‘Trigger Level’ set to the voltage crossing threshold several features become available in terms of tachometer processing.
By selecting the above options the user will be able to use the ‘Digital Panel’ display type from the signal display and thus by selecting the ‘Speed’ option in the ‘Display Type’ drop down list an RPM will be available.
“How do I balance a shaft?” seems like a fairly straightforward question, but there are a number of things that we need to understand first. Here we look at a number of key concepts that need to be understood in order perform balancing.
What does balance mean?
Well broadly speaking to balance a shaft, mass must be added or removed at certain angles. The concept being that the centre of gravity and rotational centre of the shaft will be equal when the shaft is balanced. Continue reading How do I balance a shaft?
This note describes the importance and setup of a keyphasor or tachometer signal for PROTOR Remote Monitoring Data Acquisition System (RMDAS) units. RMDAS units are provided with manual tachometer conditioning control. The note describes the control functions and there effect on the conditioned tachometer signal, it also describes the relationship between phase measurement and keyphasor triggering.